Three major halitosis components are hydrogen sulfide, methyl mercaptan, and dimethyl sulfide. The OralChroma measures Volatile Sulfide Components (VSC) in the breath and separates them into the above three components. About 1000 sets of OralChroma have been supplied to dentists, universities, and laboratories in the world since 2002.
Since the OralChroma uses gas chromatography technology, causes of halitosis can be precisely analyzed to confirm breath odor diagnostics treatment.
You can freely select where to place the OralChroma without being worried about permanent installation.
|Insert the supplied syringe upto the stopper into the oral cavity and hold the syringe between the front teeth.|
|Close mouth tightly.|
|Wait 30 seconds to one minute to have enough oral gas.|
|Slowly pull the plunger, push it in and pull it out again before removing the syringe from the mouth.|
|Wipe the top of the syringe with a tissue. Eject the remaining gas until it reaches 1.0ml (cc) by pushing the plunge.|
|Inject the remaining oral gas into the inlet of the OralChroma. The measurement will start automatically.|
DataManager is a graphical software providing halitosis strength level, patient's data history, etc.
|Model||CHM-2 (Blue: CHM-2-BL, Pink: CHM-2-PK)|
|Detection method||Simple gas chromatography using semiconductor gas sensor|
|Detection gas||Following components included in volatile sulfur compounds:
|Measuring time||4 minutes|
|Measuring unit||ng/10ml and ppb|
|Sampling method||Manual sampling with a syringe|
|Operating humidity||80% RH or less (no condensation)|
|Operating temperature||10 to 30℃|
|Storage humidity||20 to 90% RH (no condensation)）|
|Storage temperature||-20 to 60℃|
|Power supply||100 to 240V AC、50/60Hz|
* OralChroma is a measuring device.